AGILENT 34970A USB DRIVER
Update on the cable: Message 1 of Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Legal Privacy Terms Trademark Acknowledgements. Message 6 of Accepted by topic author AnthonyMN.
|Date Added:||21 December 2017|
|File Size:||24.16 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Bob Schor 28 August Most Active Software Boards: RS Communication with Agilent A. See search results instead: I’m not sure exactly what happened that manually throwing SCPI commands zgilent it wasn’t working out, but it’s working now, so I guess I’m on to debugging the program now.
How to Buy or Rent.
Agilent A and Windows 10 | Keysight Community
No matter how small the script, or how long, it always gives the same error on the last line. Authorized Sales Partners – Check availability.
Message 10 of What about your aagilent settings? Message 8 of To ensure you are purchasing an authentic Keysight product with the quality and performance you expect, purchase through Keysight authorized distribution partners. Here is a link to the connector I am using, which I believe should be right.
Solved: RS Communication with Agilent A – Discussion Forums – National Instruments
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Both settings on the DAQ and in NI MAX match up, and I checked the interface but shorting he interface pin 2 to 3 to check if it’s reading wrong, so that’s working correctly. Update on the cable: Message 7 of Thank you so much for your continued help and patience on this! Our most popular module A features a built-in thermocouple reference and 20 two-wire channels.
34970A Data Acquisition / Data Logger Switch Unit
Message 4 of Message 5 of Eventually it generates an error on the instrument, which the guidance in the manual says ” Invalid Separator” even if I’m putting in the command SYST: None of them seem to fix the issue. Has anybody ever encountered this sort of error before? I built a small program here to read the output of one channel this was an example script that they listed under one of the options.
Setup and make measurements quickly, export data or log and view real-time measurements with a variety of display types to interpret your data faster. I really don’t know what else to recomend